INSTRUMENT WFC3
CAMERA UV
# Updated +1 and -1 order calibration based on WR14 observations
# NP 10-2015

SCIENCE_EXT SCI ; Science extension
ERRORS_EXT ERR  ; Error extension
OPTKEY1 CCDCHIP
OPTVAL1 2
FFNAME None

RDNOISE 3.30 
EXPTIME EXPTIME
POBJSIZE 0.75
SMFACTOR 1.0



#
# +1st order (BEAM A) *************************************
#
BEAMA -400 215
MMAG_EXTRACT_A 35
MMAG_MARK_A 35

#
# Trace description
#
DYDX_ORDER_A 6
DYDX_A_0 165.8292581254822  -0.0017430941231037314  0.004190967371359163  7.59235690687922e-8  -4.799279736489624e-8  -2.2004234234105673e-7
DYDX_A_1 0.011329620886157242  -3.7130183285001537e-6  -5.617240044436743e-6  1.443350486490394e-10  4.682982445817497e-10  -2.6447650909722543e-10
DYDX_A_2 0.00035264609795120246  -7.408594874623914e-9  -2.9116571386002652e-8  2.0097801375431584e-12  -2.553651735835842e-12  1.151679979712031e-11
DYDX_A_3 2.441564618087794e-6  1.0837407637355353e-11  -1.1038671416977504e-10  1.0010953591602442e-14  -1.6198656708892635e-14  3.585083189192435e-14
DYDX_A_4 1.0657239033559628e-8  -1.1978178096619563e-13  5.257325236551057e-13  -5.2428993368408627e-17  1.2585855469454978e-16  -3.5340045042355594e-16
DYDX_A_5 2.3254146148411666e-11  -1.0819429215096503e-15  2.7207846994872573e-15  -3.680893302637254e-19  8.813537319288154e-19  -1.7578924823550215e-18
DYDX_A_6 1.961065359285038e-14  -1.5436874196262055e-18  2.9796117773012048e-18  -5.140747607914649e-22  1.2390045684537149e-21  -2.089896211472153e-21

#
# X and Y Offsets
#
XOFF_A -150
YOFF_A 0.0

#
# Dispersion solution
#
DISP_ORDER_A 4
DLDP_A_0 3569.985303781074  0.05037348946990385  -0.019729638983668776
DLDP_A_1 -14.01123606786801  -0.00026605332024976327  0.000143453628992614
DLDP_A_2 0.00532475340715845  -2.8506004465458117e-7  -1.0220963845208916e-7
DLDP_A_3 0.00002832445054453313  2.2356093628019544e-10  -4.23231216744567e-10
DLDP_A_4 4.7628507125190326e-8  2.9618253310343713e-12  -5.436895839636378e-13
SENSITIVITY_A wfc3_abscal_UVg280_1st_sensTV3.fits

#
# -1st order (BEAM B) *************************************
#
BEAMB -60 560
MMAG_EXTRACT_B 35
MMAG_MARK_B 35

#
# Trace description
#
DYDX_ORDER_B 6
DYDX_B_0 152.11429830237952  -0.0022862552020068557  0.002274662249920308  8.072502045187618e-8  4.453373655067323e-8  -3.3094365838784713e-7
DYDX_B_1 -0.16746161634806486  -0.000017824117269031726  2.72972894868618e-7  6.31593728694673e-9  -5.14866092097925e-10  -3.907970452627599e-9
DYDX_B_2 0.001717390508845745  -5.135728903825813e-8  -2.2385797776241416e-7  -2.5811617259434942e-11  -6.890287043425549e-12  1.7116449562705468e-10
DYDX_B_3 -0.000018438850160316978  7.843329756232926e-9  1.60472441176438e-9  -1.6476247587062665e-12  3.3205776274835525e-13  -1.751354822952019e-12
DYDX_B_4 1.0682732198064177e-7  -7.68575185796111e-11  -3.0846554347864725e-12  1.849073743906215e-14  -2.343988785559118e-15  7.124679007238576e-15
DYDX_B_5 -3.007001096601402e-10  2.768320045849178e-13  -3.1871582190076373e-15  -7.063948023352568e-17  6.101837538926873e-18  -1.2141929853655968e-17
DYDX_B_6 3.297778161402249e-13  -3.497025519744447e-16  9.585670385375236e-18  9.277242571222464e-20  -5.329274605957512e-21  7.132195335787529e-21

#
# X and Y Offsets
#
XOFF_B 300
YOFF_B 0

#
# Dispersion solution
#
DISP_ORDER_B 4
DLDP_B_0 2922.080437251009  0.04561954461101349  -0.04601844677589647
DLDP_B_1 15.408739103272021  0.00014214831922918327  -0.0001381880477541732
DLDP_B_2 -0.004050070853457972  2.4420427552298257e-7  2.1566951837861292e-7
DLDP_B_3 0.000011364871371530895  -5.306338820129571e-10  -1.9566233251610045e-9
DLDP_B_4 -1.2691664599528713e-8  7.992180218683699e-13  3.96330713195308e-12

SENSITIVITY_B wfc3_abscal_UVg280_-1st_sensTV3.fits





